The following information will help you determine where to setup your ForceDecks.
ForceDecks plates can be mounted on:
- Steel bases;
- Set into weightlifting platforms; or
- Used freestanding on any flat, hard surface
ForceDecks are designed to be mobile, however it is important ForceDecks are mounted on a surface that is:
- As rigid as possible; and
- As flat as possible;
For example, concrete floors are ideal, whereas sprung floors are not suitable.
Rebounding vibrations or “ringing” may occur if the platform is mounted on the wooden section of some Olympic lifting platforms.
If you are unsure about the rigidity of the floor you can check for vibrations by examining a force-time trace or by sending a recording to email@example.com for assessment.
We also recommend using a spirit level to identify an even part of your floor and marking out the space if you are planning to move the plates.
If embedding the force plates into the ground or into a raised lifting platform, it is important to leave adequate space around the plates to prevent contact with the surrounding walls and to allow access to cables.
Crosstalk is the unwanted transfer of force between plates. This can occur when two plates are in contact, potentially resulting in inaccurate readings.
The ForceDecks system consists of a ‘Primary’ plate and a ‘Secondary’ plate.
It is important to ensure cables between the plates and power source are routed to prevent them being stepped on or caught underneath the plate, as this can cause instability, potentially affect measurements and/or damage the cables.